Pris 119 990 kr (Ordinarie pris 189 990 kr)
- 35 % kontant vid beställning
- Resterande belopp kan vi dela upp till 5 år (60 månader), om du inte har betalningsanmärkningar.
Ring så berättar vi mer om teknisk specifikation och performance.
Tel: 0700 160 619
Traditional laser or IPL technologies employ short impulses of approximately 2-300 milliseconds, applying a vast amount of energy (12-120 J/cm2). Energy is transported to the hair root through melanin, where warmth of 65-72° Celsius is produced. Energy reaches the root of the hair follicle solely through melanin. Skin and red blood cells have a similar absorption coefficient as melanin and therefore also absorb the high levels of energy produced by the laser and IPL methods.
SHR technology, on the other hand, utilizes the melanin path only partially (50%)., and combining In-Motion technology, gently warms the skin helping penetrate down to the follicles which produce hair growth.
Research has shown that a slower, but longer heating process is considerably more effective for
permanent hair removal than high and short levels of energy. Therefore, when using SHR, the device is passed over the tissue multiple times (in motion) using low energy but a high rate of repetition (up to 10Hz, i.e. 10 times per second) instead of using the traditional method with single, high-energy impulses. Thus, the hair melanin, as well as the tissue of the stem cells, is heated with low energy at a slow pace and over a longer period of time to a comfortable temperature of 45° Celsius.
1） Open the machine,and make the water circulation in a good condition.
2）Scrape the hair and leave about 1mm long.
3）Clean the treatment area
4）Enter into the main working interface, and adjust the treatment parameters according to skin type and hair color.
5）Smare cold gel to the skin, treatment head close to the skin surface of 1 to 2 mm.for sensitive parts, the gap is about 3mm, such as the lower eyelids, lips.
6）The operator wear protection eyeglasses and the
client wear goggles.
7）Press the “Simmer”, and shot the handpiece 2-3 shoots on the floor. If a patient is the first time to have treatment, the operator need to test two to three spot behind the patient ear, and observe its response, adjust the scale and energy.
8）1 ～2 J/cm2;The treatment direction guide: from bottom to top, from outside to inside, first facial, perioral sequence of operations extended from the forehead to the nose. The degree of overlap of each spot is a spot size of 1/3.
Before do Eye, forehead operation, should be adjust the energy before operation.
When the treatment site no obvious skin reaction, the operator can increase energy density, but not exceed.
1 to 2 J/cm2 energy.
9）Treatment is over, clean the handle head and